physics-based image enhancement for infrared thermography modern infrared thermography uses...
Post on 29-Oct-2019
Embed Size (px)
Physics-based image enhancement for infrared
Stephen D. Holland∗,a, Jeremy Renshawb
aDepartment of Aerospace Engineering and Center for Nondestructive Evaluation Iowa State University
1915 Scholl Road Ames, IA 50011
bDepartment of Materials Science and Engineering and Center for Nondestructive Evaluation
Iowa State University Ames, IA 50011
Thermal imaging with an infrared camera can be used to view the location and intensity of heat sources in space and time. In a thermal conductor, thermal diffusion blurs out those heat sources. Knowledge of the physics of thermal diffusion can be used to enhance the spatial and temporal resolution of thermal images. In two dimensions, quantitative reconstruction of the heat source intensity is possible. The same algorithm applied to three-dimensional heat flows provides dramatic improvements in temporal and spatial resolution of the thermal images. Performance is illustrated both in theory and by experiment. An application example demonstrates utility to nondestructive evaluation.
Key words: Flash Thermography, Vibrothermography, Sonic Infrared, Sonic IR, image processing, image enhancement, Laplacian filter
∗Corresponding author Email addresses: email@example.com (Stephen D. Holland), firstname.lastname@example.org
Preprint submitted to NDT&E International July 11, 2011
Thermographic Nondestructive Evaluation (NDE) finds flaws or deter- mines material properties through infrared measurement of heat patterns or temperature changes. Modern infrared thermography uses real-time video/digital imaging through semiconductor focal plane arrays or microbolometer arrays. Given a blackbody – an object with perfect surface infrared emissivity – a thermal imaging camera can take accurate remote temperature measure- ments. Flash thermography involves excitation of a surface with a flash (im- pulse) of thermal energy, then monitoring surface cooling or thermal trans- mission . Established algorithms  can be used to measure thickness or identify delaminations in composite materials. Spatial resolution enhance- ment through methods such as inverse scattering  have been in use for many years to enhance the resolution of flash thermography images. Vi- brothermography, also known as “thermosonics” and “sonic infrared”, is an emerging inspection technology that finds cracks and delaminations by ob- serving heat generated from frictional rubbing of vibrating crack surfaces . Typically a high power sonic or ultrasonic transducer vibrates the specimen for approximately one second. The heating of the crack and the diffusion of the heat are observed with the infrared camera.
For the most part, infrared images are treated and processed similarly to visible light images. Standard denoising and sharpening algorithms are com- monly used with infrared images, although these algorithms were originally intended to compensate for optical blurring, not the thermal diffusion preva- lent in infrared images. There are also common processing and analysis steps that have been modified specifically for thermal images , including contrast evaluation, wavelet analysis, and principal component analysis. The quanti- tative nature of the infrared image combined with the known physics of heat conduction create the opportunity to develop processing algorithms that op- erate on physically meaningful quantities and take advantage of knowledge of heat conduction. In flash thermography, algorithms based primarily on one-dimensional heat diffusion  are often used to enhance image quality and estimate flaw depths.
There are various methods and algorithms discussed in the literature for inversion of thermal data. In general, inversion of thermal diffusion is ill- posed because of the diffusive nature of heat conduction. Inversion techniques for flash thermography attempt to estimate thickness from the conduction- driven cool-down profile of a surface exposed to a thermal impulse. The
approach of Shepard et al.  uses the time when 1D thermal diffusion slows, estimated from the inflection point of a log temperature vs. log time graph, to estimate thickness. The present work focuses on a different problem: Enhancement of lateral resolution of a surface or buried heat source. Several authors have reported analyses of flash thermography data using 3D heat- diffusion simulation [7, 8, 9] to improve resolution. Le Niliot  reports on a boundary element-based method for identifying the time-dependence of line heat sources when their position is known. Rainiere and Pagliarini  report on a Wiener filter-based method for determining the spatial distribution of the thermal diffusivity α. Spatial deconvolution of point spread functions is an established method , to improve the resolution of thermal images of subsurface sources.
In this paper, we report on a simple but effective algorithm based on the heat conduction equation for enhancement and analysis of infrared thermal images. While our method determines the heat source intensity, it does so by observation rather than inversion of the heat diffusion process. As such, it avoids the inherent difficulties in inverting diffusion.
Heat conduction in a solid is governed by the heat conduction (diffusion) equation,
∂x2 + ∂2T
∂y2 + ∂2T
k g(x, y, z, t) =
where T is temperature, k is thermal conductivity, g(x, y, z, t) is the volu- metric heat source distribution, and α is the thermal diffusivity .
Consider the two-dimensional case of a thin sheet or plate. The heat conduction equation reduces to
∂x2 + ∂2T
k g(x, y, t) =
∂t . (2)
At the surface of a specimen, the spatial (x, y) and temporal temperature distribution can be recorded with an infrared camera. The partial derivatives can be directly measured from such a sequence of calibrated infrared images of the surface of the sheet or plate. Given known or estimated thermal conductivity k, diffusivity α, and a measured sequence of infrared images,
the heat source distribution g(x, y, t) can be measured by solving Eq. 2,
g(x, y, t) = k
∂t − k
∂x2 + ∂2T
) , (3)
where k/α is equivalent to the density ρ times the specific heat cP . Eq. 3 provides a means to evaluate the spatial and temporal heat source
distribution of a two-dimensional sheet or plate from infrared images of the surface temperature. In the time-independent case (∂T/∂t = 0), this is essen- tially a Laplacian method ,. Eq. 3 uses the known thermal conduction properties of the sheet or plate to subtract out diffusion effects. Only the heat source intensity g(x, y, t), measured in Watts/meter3, remains. Equa- tion 3 is an exact solution that allows quantitative measurement of the two- dimensional heat source intensity from infrared images, and offers the poten- tial to dramatically increase the resolution of thermal images by eliminating the blurring effect of thermal diffusion.
Unfortunately no similar solution exists for heat sources in a three-dimensional solid. An infrared camera can only measure surface temperatures and thus the depth diffusion term, ∂2T/∂z2 in Eq. 1 is not measurable. Equation 3 can still be applied to the surface, but it is no longer exact nor physically quantitative. Nevertheless, both theory and experiment demonstrate that Eq. 3 can provide dramatic improvements in the spatial and temporal res- olution of thermal images even when heat flows are three dimensional. We will call the result of applying Eq. 3 to a surface image of heat flow in a three-dimensional solid the heat source estimate g̃(x, y, t).
To predict the characteristics of Eq. 3 in a three-dimensional solid, we start with the well-known Green’s function solution , and apply the method of images to determine the temperature field of a point impulse heat source at depth zs below a free surface at (xs, ys) = (0, 0) and ts = 0,
T (x, y, z, t) = 1
2 + y2 + (z − zs)2
) + exp
2 + y2 + (z + zs) 2
(4) where u(t) represents the Heaviside unit step. Eq. 4 is the Green’s function (impulse response) for thermal conduction of an impulse heat source at a distance zs below the free surface z = 0.
By applying Eq. 3 to the Green’s function temperature distribution, Eq. 4, we can evaluate the performance of the heat source estimate evaluated from Eq. 3 in eliminating the spatial and temporal spreading caused by thermal
diffusion. Substitution of Eq. 4 into Eq. 3 at the z = 0 free surface (via Eq. 1 and the second spatial derivative of Eq. 4) yields the heat source estimate g̃(x, y, t) for an impulse source,
g̃(x, y, z = 0, t) = k ∂2T
2αt − 1
) Tu(t), (5)
where T references the Green’s function temperature distribution (Eq. 4) for a heat source a distance zs below the surface. In summary, the source distribution g̃(x, y, t) of Eq. 5 is equivalent to the processed image sequence resulting from applying Eq. 3 to images of the tempe